XenoWorks™ Analog Microinjector has been designed with ease of troubleshooting in mind. While ﬂuid-ﬁlled hydraulic microinjectors are the most responsive method of manipulating cells and cellular components, over time they can develop leaks and air bubbles which need to be sealed or purged.
The design of the Analog Microinjector allows the tubing to be easily separated from the syringe and main injector body and stripped to its components; tubing and connectors. Parts such as spare tubing are readily available from Sutter (spare tubing and ﬁttings are supplied with each injector) and are quick and simple to ﬁt using commonly available Luer-lock and ferrule connectors.
Tubing length can be altered according to users' needs and ancillary devices such as three-way valves may be ﬁtted so that, for example, a single injector can control two micropipettes (for applications such as blastomere biopsy).
|Syringe Part Number||Syringe Volume||Volume Per Revolution Course Dial||Volume Per Revolution Fine Dial|
|V001180||10 µl||162 nl||16.2 nl|
|V001181||25 µl||325 nl||32.5 nl|
|V001182||50 µl||650 nl||65.0 nl|
|V001183||100 µl||1.3 µl||130 nl|
|V001184||250 µl||2.6 µl||0.26 µl|
|V001185||500 µl||5.2 µl||0.52 µl|
|V001186||1000 µl||10.4 µl||1.04 µl|